The PM311C and PM312C Pillow Speaker Intercom
Modules are for use with the Tek-CARE®NC300II Nurse
Call System using IR319A, IR320A, IR311C or IR312C
patient stations and SF401P, SF401DIG or SF301DIG
pillow speakers. The modules provide for reliable, twoway
hands-free communication from the pillow speaker
to the master station. Calling the nurse is accomplished by
pressing the push button identified with the nurse symbol.
Call placed assurance is indicated at the patient station.
If the pillow speaker is removed from its receptacle during
communication, the call is automatically transferred to the
patient station speaker. If the pillow speaker or call cord
is removed from its receptacle when not in use, a noncancellable
cord-out call is registered at the master station.
SF401, SF311 or SF312 call cords may be used without
removing modules. Communication is automatically
transferred to the patient station speaker.
• UL® 1069 Listed (when used with IR319A, IR320A,
IR311C or IR312C patient stations)
• Provides hands-free pillow speaker intercom
• Mounts directly to patient station
Width: 5" (127 mm)
Depth: 1.875" (47.6 mm)
Height: 0.5" (12.7 mm)
Construction: PC board with quality electronic components
Mounting: Plug-in connectors mounted to patientstation
Tek-CARE®NC300II Nurse Call System
IR319A or IR320A Patient Stations
with SF401DIG or SF401P Pillow Speakers
IR311C or IR312C Patient Stations
with SF301DIG Pillow Speakers
ARCHITECTS’ AND ENGINEERS’
The pillow speaker intercom module shall be TekTone®
PM311C, PM312C, or approved equal. The PM311C is
designed for use with IR319A and IR311C single patient
stations; the PM312C is designed for use with IR320A
and IR312C dual patient stations.
When used in conjunction with TekTone®’s SF401P,
SF401DIG or SF301DIG pillow speakers, the PM311C
and PM312C shall provide two-way intercom through the
All electronic components needed for this function shall
be on a printed circuit board that is mounted to the patient
station with plug-in type connections.
The PM311C and PM312C shall be UL® 1069 Listed
(when used with IR319A, IR320A, IR311C or IR312C
This product was added to our catalog on Thursday 22 May, 2014.